If it does, then you are probably communicating correctly with the instrument – if not, then you first need to figure out how establish communication before you try to make a measurement. Create an account or sign in to comment You need to be a member in order to leave a comment Create an account Sign up for a new account in our community. Most Active Software Boards: If you’re not using the latest driver from NI the plug and play one I would get it because it seems to have the most complete and automated handling of all the possible serial port settings for the host. There you will get automatically installed an instrument driver into your functions palette.
|Date Added:||5 February 2017|
|File Size:||39.35 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Make sure under Serial settings lagview the instrument’s settings match the ones in your driver. I think this can only be done from the ‘s menus – not hp34401a labview.
Keysight Instrument Drivers | Keysight (formerly Agilent’s Electronic Measurement)
hp34401a labview Also, it seem a part of VI. For instance, it tries to handle whether or not flow hp34401a labview is supported automatically. Hi, I am new to this forum and I need little help in LV. Sign In Sign Up. Is it stand alone operate? Sign in Already have an account? Started by MouradJanuary 21, Message 9 of Agilent and HP Message 3 of That error hp34401a labview a VISA timeout, so it means your instrument is not h3p4401a and probably not getting your command string, either.
Agilent Technologies / Keysight Technologies 34401A
That is really good but problem is; it gives same hp34401a labview readings ,97 or 33,98 orThe first thing I noticed when I opened your file was that it is pointing to a LLB file in the instrument lib folder for the sub VIs – that means it must be pointing to an older version of the instrument driver – the hp34401a labview version is a NI plug and play driver.
Posted January 27, I’ve found it to be a useful troubleshooting tool. Most Active Software Boards: If you’re hp34401a labview using the latest driver from NI the plug and play one I would get it because it seems to hp34401a labview the most complete and automated handling of all the possible serial port settings for the host.
hp34401a labview These errors sound like communication errors caused by misconfigured serial port settings – if hp34401a labview start and stop bits aren’t correct the data frame will be wrong. Sign up for a new account in our community.
It is nine to meet hp34401a labview. Please let me know your advice. These cables are also called null-modem, modem-eliminator, or crossover cables.
INSTR as the resource name is working properly and that is the one you are using and you have read the HPA manual and hp344001a are sure it is configured hp34401a labview for serial communication hp34401a labview I would next suspect the cable. Register a new account. Simple code for HPA.
Posted January 21, Create an account or sign in to comment You need to be a hp34401a labview in order to leave a comment Create an account Sign up hp34401a labview a new account in our community. I have check with Aglient IO Control to access device. Message 1 of From the Agilent A Manual “Connection to a Computer or Terminal To connect the multimeter to a computer hp34401a labview terminal, you must have the proper interface cable.
Posted January 24, Thanks for your infomation. You need to be a member in order to leave a comment.